The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||1 December 2014|
|PDF File Size:||12.36 Mb|
|ePub File Size:||8.35 Mb|
|Price:||Free* [*Free Regsitration Required]|
About this Manual Memory Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test. Open Battery Compartment With the meter turned off, turn it over. Match Codes Make sure that the three- number code on the new test Each box of test strips comes with its own code chip. The code strip container matches the chip provides the meter with information such as the lot number three-number code on the and expiration date of the test strips.
Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.
Roche CoaguChek XS User Manual
There is a potential risk of infection. Gather Items Gather the following items: Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1.
Get Ready Take a test strip out of the then press M container. If the numbers are different, Close the container tightly. Identify the Target Area Find the target area on the 7. Record Result Record the result according to test strip.
Fill the Capillary Tube When the flashing test strip 4.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Then touch the capillary tube to the blood drop. Keep the tube level, and allow it to fill by capillary action. Dose the Test Strip Discard the first 4 drops of blood. Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe.
View Most Recent Result The most recent test result appears. The letters mem The CoaguChek XS Meter automatically stores up to test indicate that you are viewing a results and their dates and times in its memory.
If the memory is result in memory. Erasing Stored Test Results 2.
You cannot, however, erase individual test results. To exit Erase mode without To erase all stored results: Cleaning the Meter It is important to keep the meter clean. Clean the ss whenever Cleaning the Test Strip Guide it looks dirty or, if you prefer a regular xss, clean the meter each time you open a new box of test strips.
Allow to Dry With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can test strip guide to dry for set up as necessary. Regularly check that all segments You can access these settings by going to Setup mode.
To go to of each letter, number, and symbol appear. You may see the following error messages while using the The meter is too cold or too Turn the meter off and allow it CoaguChek XS Meter. If you see an error message, first try to warm to measure correctly. For more information, see re-insert it.
If the error the Code Chip section of message reappears, discard this manual. Test Strip Interference Solution: Repeat the test using a new test Error: If you see the another method. The CoaguChek XS PT test strip may be used for patients under a combination therapy of oral anticoagulants plus heparin injections.
Indicates ciaguchek results are displayed in Seconds. For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor. A rotation scheme was used to ensure that the same instrument was not always dosed first.
For the venous testing, blood was applied from a syringe; Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. The correlation coefficient is mwnual. The correlation was 0. Coaguchdk graph shows the mean bias for INR results; the overall mean bias was 0. The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
Wurster M, Doran T. All other trademarks are the property of their respective owners. Page of Go. Table of Contents Add to my manuals Add. Table Of Contents User Manual Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
Code Chip Code Chip 2. Integrated Quality Controls Integrated Quality Controls Coagufhek for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood coxguchek, increasing the blood flow maual the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood. Preparing For A Test 1. Performing A Test 3. Using Venous Blood 2. Advanced Features Advanced Features 3.
Page 24 For more information, see re-insert it. Page 26 Repeat the test.
Page 91 For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor. Page 92 Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Page 99 The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good.
Manul to this Manuals Your Name. Enter text from picture: By pressing ‘print’ button you will print janual current page. To print the manual completely, please, download it. Don’t show me this message again.
Got it, continue to print. Don’t have an account?